FDA 510(K) Approved - M3863A Replacement Philips FR2 / FR2+ AED Battery
- Chemistry: Li-Manganese
- Rating: 12V 4.2Ah
- Version: Non-Rechargeable
- Approval: FDA Approved
- Typical Capacity: 12 hours of ON time
- Minimum Capacity: 300 Shocks
Shelf Life: A new lithium battery pack has a shelf life of five (5) years stored at proper temperatures. Although a lithium battery may be stored for up to five (5) years, it self-discharges over time. Therefore, when the battery is eventually placed in the AED, it€™s useful life will be reduced depending on how long it €œsat on the shelf€
Recommended Replace Interval: Three (3) years, depending upon usage. (see Useful Life and Expected Capacity)
Self Testing: The non-rechargeable lithium battery pack requires less maintenance then the rechargeable SLA battery pack since it never requires charging. With the lithium battery pack installed, the Philips AED automatically tests it as part of the Daily Auto test. The AED also performs the battery test the first time the AED is turned on after a new battery has been installed.
Expected Capacity: When optimally maintained, a new lithium battery pack has a capacity of 4.2 Amp hours, which is equivalent to 12 hours of €œON€ time. It is important to know that just turning the AED on uses up battery capacity. Also, each year, battery capacity decreases while the battery is in the AED due to the normal self-discharge rate. After 4 years of absolutely no patient use, approximately 35% of the useful life will remain.
Warranty: Maximum of 4 Years shelf life. If installed: 12 Hours On or 300 shocks, which ever comes first
Ground Transport Only: Cannot be shipped by air (Priority Overnight, Standard Overnight, 2nd Day, 3rd Day, International) due to federal regulations regarding the transport of primary lithium cells on aircraft.
This Battery, Battery Posts, Terminals, and Related Accessories can expose you to chemicals including Lead, Lead Compounds and other chemicals which are known to the State of California to cause Cancer and birth Defects or other reproductive harm. Wash hands after handling. For more information go to: www.P65Warnings.ca.gov